Correction For Cold-junction Errors
: PYROMETRY AND PYROMETERS
: The Working Of Steel
The voltage generated by a thermo-couple of an electric pyrometer is
dependent on the difference in temperature between its hot junction,
inside the furnace, and the cold junction, or opposite end of the
thermo-couple to which the copper wires are connected. If the
temperature or this cold junction rises and falls, the indications
of the instrument will vary, although the hot junction in the furnace
may be at a constan
A cold-junction temperature of 75 deg.F., or 25 deg.C., is usually adopted
in commercial pyrometers, and the pointer on the pyrometer should
stand at this point on the scale when the hot junction is not heated.
If the cold-junction temperature rises about 75 deg.F., where base metal
thermo-couples are used, the pyrometer will read approximately 1 deg.
low for every 1 deg. rise in temperature above 75 deg.F. For example, if the
instrument is adjusted for a cold-junction temperature of 75 deg., and
the actual cold-junction temperature is 90 deg.F., the pyrometer will
read 15 deg. low. If, however, the cold-junction temperature falls below
75 deg.F., the pyrometer will read high instead of low, approximately
1 deg. for every 1 deg. drop in temperature below 75 deg.F.
With platinum thermo-couples, the error is approximately 1/2 deg. for
1 deg. change in temperature.